Beckman.com is conducting maintenance on its shopping cart Friday, July 10th at 9 pm PST thru Sunday, July 12th at 10 pm PST. During this time, orders can still be placed by calling Client Services. We look forward to providing you an improved shopping experience.
Most online TOC analyzers are designed to have a sample constantly flowing through them. Hence the exposure to the oxidizing UV light is dependent on how fast the water flows past the UV light. Increases or decreases in the water loop pressure can affect the speed at which the water passes through the TOC analyzer, thus affecting the time exposed to the UV lamp and hence affecting the TOC level reported by the TOC analyzer. To get ‘round this problem, some TOC analyzers have an averaging function, which make them report the average TOC level over a period of time. In this case it is possible that the TOC analyzer may not report a TOC excursion.
Oops. The page you wanted could not be found, so we brought you to something similar. If you still can't find what you want, try searching our site instead.