A scheduled system update will be performed on Friday, March 1st from 6:00 PM PST to Saturday, March 2nd 10:00 PM PST . During this timeframe, customers will not be able to place orders online. Please contact us for assistance. We appreciate your patience while we continue to improve the online experience at beckman.com.
Discover paperless reporting features with the QbD1200 laboratory TOC analyzer. Easy reporting for your point-of-use grab sample analysis.
Welcome to the third virtual demo of the QbD1200 where we cover the paperless electronic record reporting feature.
The first step is to configure your network settings. This typically will only have to be done one time and you may wish to enlist the help of your IT team.
From the home screen, go to Instrument Setup and Systems Settings and Network.
Now choose the SFTP tab and enter the IP address for the server you want to send your report to.
Choose the protocol — either FTP or FTP Secure, depending on your server — and enter the username and password for the server.
Test the connection. Note the green check. It gives you reassurance that you have a good connection. Once complete, save the settings.
Once they're saved, you should not have to adjust the network settings again.
Let's go back to the home screen. Now we're ready to send some paperless electronic reports.
From the home screen, select Reports and Data. Then choose the report type. Here we'll choose measurement data.
Now we can filter the report. In this case, we'll choose all measurement data from all dates. Choose the report type and then the output.
Here, of course, we'll use the Paperless Export.
It will ask us if we want to start transmitting the report and then show a status bar of the transmission over Ethernet. Once complete, you'll get a message telling you it's complete.
Now we can go back and choose a different report type if we want.
So we'll go back to that Select Report Type screen. Here we'll choose a calibration. The next step is to filter it.
Now we'll choose calibration data from the last month.
Once again, select the report format style in the output, selecting paperless electronic record output.
Start the transmission . . . and the transmit is complete.
If we go back to the Select Report Type step, you can see that all types of qualification routines and measurements are permanently stored on the QbD1200. The paperless reporting feature is a nice way to send these over a secure protocol to a central server which allows you to keep all auditable electronic records in a single place for easy retrieval.
This concludes the QbD1200 virtual demo featuring paperless electronic records.
No Related Videos
Bleiben Sie auf dem Laufenden mit unserem Newsletter.
Erhalten Sie aktuelle Informationen zu relevanten Themen von den Experten von Beckman Coulter Life Sciences. Geben Sie Ihre E-Mail ein.
NOT ALL PRODUCTS ARE AVAILABLE IN ALL COUNTRIES. PRODUCT AVAILABILITY AND REGULATORY STATUS DEPENDS ON COUNTRY REGISTRATION PER APPLICABLE REGULATIONS The listed regulatory status for products correspond to one of the below: IVD: In Vitro Diagnostic Products. These products are labeled "For In Vitro Diagnostic Use." ASR: Analyte Specific Reagents. These reagents are labeled "Analyte Specific Reagent. Analytical and performance characteristics are not established." CE-IVD, CE: Products intended for in vitro diagnostic use and conforming to the In Vitro Diagnostic Regulation (IVDR) (EU) 2017/746. (Note: Devices may be CE marked to other directives.) RUO: Research Use Only. These products are labeled "For Research Use Only. Not for use in diagnostic procedures." LUO: Laboratory Use Only. These products are labeled "For Laboratory Use Only." No Regulatory Status: Non-Medical Device or non-regulated articles. Not for use in diagnostic or therapeutic procedures.