Beckman.com is conducting maintenance on its shopping cart Friday, July 10th at 9 pm PST thru Sunday, July 12th at 10 pm PST. During this time, orders can still be placed by calling Client Services. We look forward to providing you an improved shopping experience.
How can I differentiate between particulate in the sample, the sample probe, or the sheath solution?
If there’s particulate in the sample probe, dep. On the rate you flow it at. So change the flow rate and you will see more or less debris. If it’s in the sheath, it will stay constant because the sheath rate does not change. If after you clean the sample probe and you still have a lot of noise, it’s most likely in the sheath. If both are clean, if you have a lot of debris in the sample, you need cleaner buffers.
Oops. The page you wanted could not be found, so we brought you to something similar. If you still can't find what you want, try searching our site instead.