Beckman.com is conducting maintenance on its shopping cart Friday, July 10th at 9 pm PST thru Sunday, July 12th at 10 pm PST. During this time, orders can still be placed by calling Client Services. We look forward to providing you an improved shopping experience.
Why does the European Pharmacopoeia suggest that TOC analyzers should completely oxidize the TOC present in the sample to CO2?
TOC analyzers calculate the amount of TOC present in the sample by oxidizing the TOC and then measuring the amount of CO2 generated. If the TOC in the sample is not completely oxidized, then the TOC analyzer will not report correctly the TOC present in the sample.
TOC analyzers work by oxidizing the TOC in the sample to turn the carbon present to CO2. If oxidation is not complete then not all of the organic carbon is turned into CO2. This could result in the TOC either over reporting, or under reporting. If the intermediate species formed as the TOC is oxidized are organic acids, then the TOC analyzer may over-report the TOC levels. The PAT700 uses dynamic end-point detection to detect during oxidation that the levels of CO2 have stopped changing indicating that all the TOC has been completely oxidized to CO2.
Oops. The page you wanted could not be found, so we brought you to something similar. If you still can't find what you want, try searching our site instead.